Accessories for Breeze Tronic Pro device
The sensor unit is located in the therapy room or in the outgoing ducting and it is connected to the main unit. It includes the following sensors:
a. Humidity sensor: The LCD display will show the humidity in the therapy room. If the humidity is out of the 40-70% range the therapy won’t be efficient, a reduced healing effect will be delivered only. In this case the main unit will inform the operator that the humidifier or dehumidifier has to be switched on.
b. Thermometer: The LCD display will show the temperature in the therapy room. If the temperature is out of the 20-25% range the therapy won’t be efficient, a reduced healing effect will be delivered only. In this case the main unit will inform the operator that the heating or air conditioning has to be switched on.
c. Laser sensors: Laser beam is sent out at the left side of the sensor unit. On the other end laser detectors are located. They measure the strength of the incoming laser beam, the frequency and length of the disruption and the direction and strength of the diverted laser beams At every start the main unit runs a calibration to have values for the clean air with 0 mg/m3 concentration.
A media box - which is located in the therapy room is connected to the main unit. It shows on a regular TV the remaining time, concentration, humidity, temperature and the medical conditions selected for that sessions to the patients.
The remote control is located at the reception optionally if the reception is far from the technical room. It can be added if the operator would not hear the beeping of the device if salt would run out or a malfunction would happen. It includes another touch screen LCD with the same information on it as on the LCD in the Main Unit. The purpose of this is to give the opportunity to the operator to start therapy sessions remotely and also to have feedback if there is a problem with the feeder, fan, humidity, etc.